Open as PDF
All display marks appear for approx. 1 sec-
ond and buzzer beeps.
After all display marks are disappeared, de-
flation mark “▼” appears for approx. 1 sec-
ond (three short beeps notify you of ready
for measurement) then the last pressure set-
ting value appears for approx. 3 seconds.
Then the “
” appears at the display.
Ready for measurement
2. Attaching of wrist cuff
• Open the wrist cuff as the figure.
• Turn left hand palm upwards and wrap the cuff around the wrist in
the left-figure way that air hose is facing the palm of the hand.
• Wrap the wrist cuff on bare skin.
- Eng 6 -